Specified Meterset (3008,0042) contains the MU as specified in the corresponding RT Plan at a given control point.
Delivered Meterset (3008,0044) shall contain one of the following three values:
the Meterset value at which the delivery of the current beam started
the Specified Meterset
the Meterset value at which the delivery of the current beam ended
Control points which already have been treated in an earlier session shall contain the Meterset value at which the delivery of the current beam started. Control points which have been completely treated during the current session shall contain the Specified Meterset value for this Control Point. Control Points which have not yet been treated or not completely shall contain the total delivered MU up to the point where the interruption has occurred (i.e. the last control point treated).
This can be expressed by the following equation:
DelMS[CP _{ n } ] = MAX ( StartMS, MIN ( SpecMS[CP _{ n } ], EndMS ))
with
DelMS[CP _{ n } ]: Delivered Meterset value at control point n
SpecMS[CP _{ n } ]: Specified Meterset value at control point n
StartMS: Meterset value where delivery of current beam started
EndMS: Meterset value where delivery of current beam ended
By this definition it is unambigously recorded, which ‘segments’ of control points have been delivered in case of partial treatments.
Example 1: 2 Control Points, 2 complete Partial Treatments
Total Meterset: 50
Interruption at: 18 MU
RT Plan:
[pic]
RT Treatment Record 1 of 2
[pic]
RT Treatment Record 2 of 2
[pic]
Example 2: 4 Control Points, 3 complete Partial Treatments
Total Meterset: 50
Interruption at: 25, 30 MU
RT Plan:
[pic]
RT Treatment Record 1 of 3
[pic]
RT Treatment Record 2 of 3
[pic]
RT Treatment Record 3 of 3
[pic]
Example 3: 7 Control Points, 2 Partial Treatments with small gap
Total Meterset: 50
Interuption at: 25 MU, Resumption at 30 MU
RT Plan:
[pic]
RT Treatment Record 1 of 2
[pic]
RT Treatment Record 2 of 2
[pic]