Table C.8-9 contains the Attributes that describe Nuclear Medicine Images.
Table C.8-9NM IMAGE MODULE ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Image Type||(0008,0008)||1||Image identification characteristics. See C.188.8.131.52.1 for specialization.|
|Image ID||(0054,0400)||3||User or equipment generated Image identifier.|
|Lossy Image Compression||(0028,2110)||1C||Specifies whether an Image has undergone lossy compression. Enumerated Values: 00 = Image has NOT been subjected to lossy compression. 01 = Image has been subjected to lossy compression. See C.184.108.40.206.5 Required if Lossy Compression has been performed on the Image.|
|Counts Accumulated||(0018,0070)||2||Sum of all gamma events for all frames in the image. See C.220.127.116.11.2 for specialization.|
|Acquisition Termination Condition||(0018,0071)||3||Description of how the data collection was stopped. Defined Terms: CNTS = counts DENS = density MANU = manual OVFL = data overflow TIME = time TRIG = physiological trigger See C.18.104.22.168.3 for specialization.|
|Table Height||(0018,1130)||3||The height of the patient table in mm. The range and values of this element are determined by the manufacturer. Should not be included if Image Type (0008,0008), Value 3, is TOMO, GATED TOMO, RECON TOMO or RECON GATED TOMO.|
|Table Traverse||(0018,1131)||3||Location of the patient table (or gantry relative to the table) in mm. The range and values of this element are determined by the manufacturer. Should not be included if Image Type (0008,0008), Value 3, is TOMO, GATED TOMO, RECON TOMO or RECON GATED TOMO.|
|Actual Frame Duration||(0018,1242)||1C||Elapsed time for data acquisition in msec. Required if Image Type (0008,0008) Value 3 is: WHOLE BODY or STATIC. See C.22.214.171.124.4 for specialization.|
|Count Rate||(0018,1243)||3||Maximum count rate achieved during the acquisition in counts/sec.|
|Processing Function||(0018,5020)||3||Code or description of processing functions applied to the data.|
|Corrected Image||(0028,0051)||3||One or more values that indicate which, if any, corrections have been applied to the image. Corrections are applied to all frames in the image. Defined Terms: UNIF = flood corrected COR = center of rotation corrected NCO = non-circular orbit corrected DECY = decay corrected ATTN = attenuation corrected SCAT = scatter corrected DTIM = dead time corrected NRGY = energy corrected LIN = linearity corrected MOTN = motion corrected CLN = count loss normalization; Any type of normalization applied to correct for count loss in Time Slots.|
|Whole Body Technique||(0018,1301)||3||The type of scan performed. Used only if Image Type (0008,0008), Value 3, contains the value WHOLE BODY. Enumerated Values: 1PS = one pass 2PS = two pass PCN = patient contour following employed MSP = multiple static frames collected into a whole body frame.|
|Scan Velocity||(0018,1300)||2C||The speed of the camera motion over the body in mm/sec. Required if Image Type (0008,0008) Value 3 contains the value WHOLE BODY.|
|Scan Length||(0018,1302)||2C||Size of the imaged area in the direction of scanning motion, in mm. Required if Image Type (0008,0008) Value 3 contains the value WHOLE BODY.|
|Trigger Source or Type||(0018,1061)||3||Text indicating trigger source. Defined Term: EKG|
|Include 'General Anatomy Optional Macro' Table 10-7||No Context ID for the Anatomic Region Sequence is defined.|
|Referenced Image Real World Value Mapping Sequence||(0040,9094)||3||A sequence of real world value mapping items. One or more Items are permitted in this sequence. Each item defines a single mapping and a list of images to which the mapping applies.|
|>Include Real World Value Mapping Macro Table C.7.6.16-12||Defined Context ID for Measurement Units Code Sequence is 84 “PET Units for Real World Value Mapping”.|
Notes: 1. Content Date (0008,0023) and Content Time (0008,0033) are included in the General Image Module, Table C.7-7, whenever the images are temporally related. For this purpose, all NM Images are considered temporally related, so that these elements are included in an NM Image.
2. Referenced Overlay Sequence (0008,1130) and Referenced Curve Sequence (0008,1145) were previously included in this Module as optional Attributes but have been retired. See PS 3.3 2004.
For NM images, Image Type (0008,0008) Value 3 is specified to be Type 1 and use one of the following Enumerated Values:
RECON GATED TOMO
For NM images, Image Type (0008,0008) Value 4 is specified to use one of the following Enumerated Values:
Note: For NM images, Image Type (0008,0008) Value 1 will be ORIGINAL for all raw data and reconstructed images. DERIVED may be appropriate for some other results images.
For NM images, Image Type (0008,0008) Value 2 will be PRIMARY.
Counts Accumulated (0018,0070) is the total of all gamma events accumulated in all frames of this Image. This attribute applies to acquisition data, and often does not apply to processed images (DERIVED, SECONDARY).
Acquisition Termination Condition (0018,0071) is the method of acquisition termination which was actually applied to the data collection. The Defined Terms and definitions are:
CNTS = preset count limit was reached
DENS = preset count density was reached
MANU = acquisition was terminated manually
OVFL = acquisition was terminated automatically by pixel data overflow condition
TIME = preset time limit was reached
TRIG = preset number of physiological triggers was reached
Actual Frame Duration (0018,1242) is defined as the elapsed time in msec for a single frame of an acquisition. For some types of multi-frame images, Actual Frame Duration (0018,1242) may have a more specialized meaning as defined in the appropriate IOD Module.