This Module applies to a TOMO Multi-frame Image. This module is present when the Image Type (0008,0008) Value 3, is equal to TOMO, GATED TOMO, RECON TOMO, or RECON GATED TOMO. The elements found in this module describe the acquisition of the Image.
Table C.8-12NM TOMO ACQUISITION MODULE ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Rotation Information Sequence||(0054,0052)||2||Sequence of Items that describe TOMO rotational groups. A new rotation is defined whenever the direction of the detector motion changes, or the Table Traverse (0018,1131) changes. Zero or more Items shall be included in this sequence. The number of items shall be equal to Number of Rotations (0054,0051). If Rotation Vector (0054,0050) is present, the first item corresponds to frames with value of 1 in the Rotation Vector (0054,0050), the second item with value 2, etc.|
|>Start Angle||(0054,0200)||1||Position of the detector about the patient for the start of this rotation, in degrees. Zero degrees is referenced to the origin at the patient's back. Viewing from the patient's feet, angle increases in a counter-clockwise direction (detector normal rotating from the patient's back towards the patient's left side).|
|>Angular Step||(0018,1144)||1||The angular scan arc step between views of the TOMO acquisition, in degrees. See C.22.214.171.124.1 for further explanation.|
|>Rotation Direction||(0018,1140)||1||Direction of rotation of the detector about the patient. See Start Angle (0054,0200) for further explanation of direction. Enumerated Values: CW = clockwise (decreasing angle) CC = counter-clockwise (increasing angle).|
|>Scan Arc||(0018,1143)||1||The effective angular range of the scan data in degrees. The value shall be positive.|
|>Actual Frame Duration||(0018,1242)||1||Nominal acquisition time per angular position, in msec.|
|>Radial Position||(0018,1142)||3||Radial distance of the detector from the center of rotation, in mm. It shall have a single value which is an average value for this rotation, or it shall have one value per angular view.|
|>Distance Source to Detector||(0018,1110)||2C||Distance in mm from transmission source to the detector face. Required if Image Type (0008,0008) Value 4 is TRANSMISSION.|
|>Number of Frames in Rotation||(0054,0053)||1||Number of angular views in this rotation.|
|>Table Traverse||(0018,1131)||3||Location of the patient table (or gantry relative to the table) in mm.The range and values of this element are determined by the manufacturer.|
|>Table Height||(0018,1130)||3||The distance in mm of the top of the patient table to the center of rotation. Table height below the center of rotation has a positive value.|
|Type of Detector Motion||(0054,0202)||3||Describes the detector motion during acquisition. Enumerated Values: STEP AND SHOOT = Interrupted motion, acquire only while stationary. CONTINUOUS = Gantry motion and acquisition are simultaneous and continuous. ACQ DURING STEP = Interrupted motion, acquisition is continuous.|
Angular Step (0018,1144) is the nominal frame-to-frame incremental angle for TOMO and GATED TOMO acquisition images, defined in degrees. The Angular Step (0018,1144) shall be a positive number. Summation of Angular Step values is not defined to give accurate Angular Position or Scan Arc values. The Angular Step is the effective angular spacing between resultant frames of the Multi-framed planar image data.