Table C.8.19.5-1 contains IOD Attributes that describe an X-Ray detector.
Table C.8.19.5-1X-RAY DETECTOR MODULE ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Include ‘Digital X-Ray Detector Macro’ Table C.8-71b|
|Physical Detector Size||(0018,9429)||1||Dimensions of the physical detector measured in mm as a row size followed by a column size.|
|Position of Isocenter Projection||(0018,9430)||1C||Position of the Isocenter projection on the detector plane measured in fractional physical detector elements as a distance along the column direction followed by a distance along the row direction from the center of the TLHC detector element of a rectangle circumscribing the physical detector area. Required if Isocenter Reference System Sequence (0018,9462) is present.|