The XA/XRF X-Ray IODs use the General Series module described in section C.7.3.1, specialized by the XA/XRF Series Module, to describe the DICOM Series Entity specified in A.47 and A.48. It is defining what constitutes a Series for the context of projection XA/XRF device.
Table C.8.19.1-1 specifies the Attributes that identify and describe general information about the XA/XRF Series.
Table C.8.19.1-1XA/XRF SERIES MODULE ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Modality||(0008,0060)||1||Type of equipment that originally acquired the data used to create the images in this Series. Enumerated Values: XA RF See section C.184.108.40.206.1 for further explanation.|
|Series Number||(0020,0011)||1||A number that identifies this Series.|
|Referenced Performed Procedure Step Sequence||(0008,1111)||1C||Uniquely identifies the Performed Procedure Step SOP Instance to which the Series is related (e.g. a Modality or General-Purpose Performed Procedure Step SOP Instance). Only a single Item shall be included in this sequence. Required if the Modality Performed Procedure Step SOP Class, General Purpose Performed Procedure Step SOP Class is supported.|
|>Include ‘SOP Instance Reference Macro' Table 10-11|