The MR IODs use the General Series module described in section C.7.3.1, specialized by the MR Series Module, to describe the DICOM Series Entity described in A.1.2.3, and to define what constitutes a Series for the context of MR device.
Table C.8-101 specifies the Attributes that identify and describe general information about the MR Series.
Table C.8-101MR SERIES MODULE ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Modality||(0008,0060)||1||Type of equipment that originally acquired the data used to create the images in this Series. Enumerated Values: MR See section C.184.108.40.206.1 for further explanation.|
|Referenced Performed Procedure Step Sequence||(0008,1111)||1C||Uniquely identifies the Performed Procedure Step SOP Instance to which the Series is related (e.g. a Modality or General-Purpose Performed Procedure Step SOP Instance). Only a single Item shall be included in this sequence. Required if the Modality Performed Procedure Step SOP Class , General Purpose Performed Procedure Step SOP Class is supported.|
|>Include ‘SOP Instance Reference Macro' Table 10-11|