Table C.7.6.16-4 specifies the attributes of the Plane Position (Patient) Functional Group macro.
Table C.7.6.16-4PLANE POSITION (PATIENT) MACRO ATTRIBUTES
|Attribute Name||Tag||Type||Attribute Description|
|Plane Position Sequence||(0020,9113)||1||Identifies the position of the plane of this frame. Only a single Item shall be included in this sequence.|
|>Image Position (Patient)||(0020,0032)||1C||The x, y, and z coordinates of the upper left hand corner (center of the first voxel transmitted) of the frame, in mm. See C.126.96.36.199.1 and C.188.8.131.52.3.1 for further explanation. Note: In the case of CT images with an Acquisition Type (0018,9302) of CONSTANT_ANGLE the image plane is defined to pass through the data collection center and be normal to the central ray of the diverging X-Ray beam. Required if Frame Type (0008,9007) Value 1 of this frame is ORIGINAL and Volumetric Properties (0008,9206) of this frame is other than DISTORTED, may be present otherwise.|
C.184.108.40.206.3.1 Position and Orientation for SAMPLED Frames
In the case of Volumetric Properties (0008,9206) having a value of SAMPLED, the Image Position (0020,0032), Image Orientation (0020,0037) and Slice Thickness (0018,0050) shall represent the volume from which the frame was derived based on the orientation of the sampling performed.
Note: For example in the case of MAX_IP:
The Image Orientation shall be the direction of the ray used for projection of the center of the plane.
The image position shall contain the x, y, and z coordinates of the intersection of the mid-plane of the sampled volume with the ray used to project the upper left hand corner of the frame.
The Slice Thickness shall contain the distance that the ray used for projection of the center of the plane traveled through the volume.