Table A.33.1-1 Grayscale Softcopy Presentation State IOD MODULES
|Clinical Trial Subject||C.7.1.3||U|
|Clinical Trial Study||C.7.2.3||U|
|Clinical Trial Series||C.7.3.2||U|
|Presentation||Presentation State Identification||C.11.10||M|
|State||Presentation State Relationship||C.11.11||M|
|Presentation State Shutter||C.11.12||M|
|Presentation State Mask||C.11.13||M|
|Mask||C.7.6.10||C - Required if the referenced image(s) are multi-frame and are to be subtracted|
|Display Shutter||C.7.6.11||C - Required if a Display Shutter is to be applied to referenced image(s) and the Bitmap Display Shutter Module is not present|
|Bitmap Display Shutter||C.7.6.15||C - Required if a Display Shutter is to be applied to referenced image(s) and the Display Shutter Module is not present|
|Overlay Plane||C.9.2||C - Required if Overlay is to be applied to referenced image(s) or the Bitmap Display Shutter Module is present|
|Overlay Activation||C.11.7||C- Required if referenced image contains overlay data that is to be displayed or Presentation State Instance contains Overlay data other than Bitmap Shutter|
|Graphic Annotation||C.10.5||C - Required if Graphic Annotations are to be applied to referenced image(s)|
|Spatial Transformation||C.10.6||C - Required if rotation or flipping are to be applied to referenced image(s)|
|Graphic Layer||C.10.7||C - Required if Graphic Annotations or Overlays or Curves are to be applied to referenced image(s)|
|Modality LUT||C.11.1||C - Required if a Modality LUT is to be applied to referenced image(s)|
|Softcopy VOI LUT||C.11.8||C - Required if a VOI LUT is to be applied to referenced image(s)|
|Softcopy Presentation LUT||C.11.6||M|
In the Grayscale Softcopy Presentation State IOD, the Presentation Series Module specializes some Attributes of the General Series Module, the Presentation State Mask Module specializes some Attributes of the Mask Module, and the Presentation State Shutter Module specializes some Attributes of the Bitmap Display Shutter and Display Shutter Modules.
Notes: 1. Subtraction between different images is not supported.
2. The Mask Module condition implies that it need not be supported by an SCP that supports presentation states only for single frame image storage SOP Classes, or instances of multi-frame image Storage SOP Classes that contain only one frame.
3. The Display Shutter may be used to darken image areas that surround important information and exclude extraneous bright areas that increase glare and ambient lighting impairing image interpretation. For example, unexposed areas in a CR image might be obscured using the Display Shutter, rather than permanently replacing image pixels in those areas.
4. This IOD does not support the storage of a multi-frame overlay in the IOD itself, but does support selective activation of multi-frame overlays within the referenced images via the Overlay Activation Module.