This template collects an individual impression the mammography CAD device had for a lesion, non-lesion object, or correlation of related objects. The details of the impression and recommendation are expressed in the Mammography CAD Impression/Recommendation Body (see TID 4002). The data from which the details are inferred are expressed in the Composite Features (see TID 4004) and/or Single Image Findings (see TID 4006) of which there may be several.
The sub-tree headed by this template is illustrated in PS 3.17 in the Annex on Mammography CAD SR Content Tree Structure.
TID 4003MAMMOGRAPHY CAD INDIVIDUAL IMPRESSION/RECOMMENDATIONType: Non-Extensible Order: Significant
|NL||Rel with Parent||VT||Concept Name||VM||Req Type||Condition||Value Set Constraint|
|1||CONTAINER||EV (111034, DCM, “Individual Impression/ Recommendation”)||1||M|
|2||>||HAS CONCEPT MOD||CODE||EV (111056, DCM, “Rendering Intent”)||1||M||DCID (6034) Intended Use of CAD Output|
|3||>||CONTAINS||INCLUDE||DTID (4002) Mammography CAD Impression / Recommendation Body||1||U|
|4||>||CONTAINS||INCLUDE||DTID (4004) Mammography CAD Composite Feature||1-n||MC||At least one of rows 4, 5 shall be present.|
|5||>||CONTAINS||INCLUDE||DTID (4006) Mammography CAD Single Image Finding||1-n||MC||At least one of rows 4, 5 shall be present.|
Content Item Descriptions
|Rendering Intent||This content item constrains the SCP receiving the Mammography CAD SR IOD in its use of the contents of this template and its target content items. Mammography CAD devices may opt to use data marked “Not for Presentation” or “Presentation Optional” as input to subsequent mammography CAD processing steps. Refer to PS 3.4, Annex O Structured Reporting Standard SOP Classes for SCU and SCP Behavior.|