This general template provides detailed information on projection X-Ray dose value accumulations over several irradiation events from the same equipment (typically a study or a performed procedure step).

TID 10004
*
ACCUMULATED PROJECTION X-RAY DOSE
*
*
Type: Extensible Order: Significant
*

NL | Rel with Parent | VT | Concept Name | VM | Req Type | Condition | Value Set Constraint | |

1 | NUM | EV (113722, DCM, “Dose Area Product Total”) | 1 | M | Units = EV (Gy.m2, UCUM, “Gy.m2”) | |||

2 | NUM | EV (113725, DCM, “Dose (RP) Total”) | 1 | MC | IF any of the values of TID (10001) Row 14 are not (113858, DCM, “MPPS Content”). May be present otherwise. | Units = EV (Gy, UCUM, “Gy”) | ||

3 | NUM | EV (113726, DCM, “Fluoro Dose Area Product Total”) | 1 | MC | IFF TID(10003) Row 4 value = (P5-06000, SRT, “Fluoroscopy”) for at least one irradiation event | Units = EV (Gy.m2, UCUM, “Gy.m2”) | ||

4 | NUM | EV (113728, DCM, “Fluoro Dose (RP) Total”) | 1 | MC | IFF TID(10003) Row 4 value = (P5-06000, SRT, “Fluoroscopy”) for at least one irradiation event AND any of the values of TID (10001) Row 14 are not (113858, DCM, “MPPS Content”). | Units = EV (Gy, UCUM, “Gy”) | ||

5 | NUM | EV (113730, DCM, “Total Fluoro Time”) | 1 | MC | IFF TID(10003) Row 4 value = (P5-06000, SRT, “Fluoroscopy”) for at least one irradiation event. | Units = EV (s, UCUM, “s”) | ||

6 | NUM | EV (113727, DCM, “Acquisition Dose Area Product Total”) | 1 | M | Units = EV (Gy.m2, UCUM, “Gy.m2”) | |||

7 | NUM | EV (113729, DCM, “Acquisition Dose (RP) Total”) | 1 | MC | IF any of the values of TID (10001) Row 14 are not (113858, DCM, “MPPS Content”). May be present otherwise. | Units = EV (Gy, UCUM, “Gy”) | ||

8 | NUM | EV (113855, DCM, “Total Acquisition Time”) | 1 | M | Units = EV (s, UCUM, “s”) | |||

9 | NUM | EV (113731, DCM, “Total Number of Radiographic Frames”) | 1 | U | Units = EV (1, UCUM, “no units”) | |||

10 | CODE | EV (113780, DCM, “Reference Point Definition”) | 1 | MC | IF Row 2, Row 4 or Row 7 is present and Row 11 is not present. | DCID (10025) Radiation Dose Reference Points | ||

11 | TEXT | EV (113780, DCM, “Reference Point Definition”) | 1 | MC | IF Row 2, Row 4 or Row 7 is present and Row 10 is not present. |

*
Content Item Descriptions
*

Row 1 | Sum of acquisition and fluoroscopy |

Row 2 | Sum of acquisition and fluoroscopy, relative to reference point. |

Rows 3-5 | Fluoroscopic component only |

Rows 6-8 | Acquisition component only |

Row 10 | A coded definition of the Reference Point (RP) used for RP-related dose values. |

Row 11 | A text definition of the Reference Point (RP) used for RP-related dose values. |